Cookies ussage consent
Our site saves small pieces of text information (cookies) on your device in order to deliver better content and for statistical purposes. You can disable the usage of cookies by changing the settings of your browser. By browsing our site without changing the browser settings you grant us permission to store that information on your device.I agree, do not show this message again.
J. NEDOMA1,* , S. KEPAK1, J. CUBIK1, J. FRNDA2, M. DURICA2, M. FAJKUS1, R. MARTINEK3
- Department of Telecommunications, Faculty of Electrical Engineering and Computer Science, VSB - Technical University of Ostrava, 17 listopadu 15/2172, 708 33 Ostrava, Czech Republic
- Department of Quantitative Methods and Economic Informatics, Faculty of Operation and Economics of Transport and Communications, University of Zilina, Univerzitna 1, 010 26 Zilina, Slovakia
- Department of Cybernetics and Biomedical Engineering, Faculty of Electrical Engineering and Computer Science, VSB - Technical University of Ostrava, 17. listopadu 15/2172, 708 33 Ostrava, Czech Repub
This paper deals with the description of S1-S1 interval detection of the phonocardiographic signal obtained by a fiber-optic Mach-Zehnder interferometer. A non-invasive measuring probe encapsulated in a bicomponent (base and catalyst) entitled Zhermack ZA 50 LT was designed. The probe is immune to the electromagnetic interferences (EMI) and therefore can be operated under strong magnetic fields. The laboratory results (obtained from 8 volunteers) were compared to the reference electrocardiograph (ECG) signal and evaluated by an objective Bland-Altman (B-A) method. Based on the results, it is possible to say that the S1-S1 interval detection with the presented fiber-optic interferometric probe is as reliable as the measurement of time between the R-R wave of the ECG signal since the B-A analysis resulted in 95.59 % samples laid in ± 1.96 SD. The test carried out in a real Magnetic Resonance Imaging (MRI) environment demonstrated that the presence of the probe in the MRI scanner does not affect the quality of this imaging modality. The developed sensor can be suitable for cardiac triggering..
Fiber-optic sensor, Interferometer, Physiological parameters monitoring.
Submitted at: May 16, 2019
Accepted at: Dec. 10, 2019
J. NEDOMA, S. KEPAK, J. CUBIK, J. FRNDA, M. DURICA, M. FAJKUS, R. MARTINEK, Vital sign monitoring: a practical solution by a MR compatible phonocardiography interferometric probe, Journal of Optoelectronics and Advanced Materials Vol. 21, Iss. 11-12, pp. 656-662 (2019)
- Download Fulltext
- Downloads: 86 (from 70 distinct Internet Addresses ).